e-corder can record dissolved carbon dioxide and/or ammonia concentrations with a suitable gas sensing electrodes. These typically work by allowing carbon dioxide (or ammonia) to diffuse across a permeable plastic membrane into the inner compartment of the electrode where they reversibly react with a buffer solution and in doing so alter the pH. The electrode senses the change in pH of its internal solution and this signal change is calibrated in suitable units of CO2 (or NH3) concentration. Carbon dioxide electrodes are also used to estimate concentrations of bicarbonate and carbonate.
These electrodes can be used with:
Electrodes can be obtained from:
- Microelectrodes Inc. see the miniature MI-720 and MI-740 electrodes (or corresponding 'Flow Thru' models 8-720, 16-720, 8-740 or 16-740). These electrodes have an internal Ag/AgCl reference. Microelectrodes will fit these electrodes with BNC connectors on request.
- Thermo Scientific offer 12 mm OD sized electrodes
DISCLAIMER: The appearance of a third party company or product on this page is for informational purposes and constitutes neither a recommendation nor an endorsement.