Troubleshooting Connecting Front-End Amplifiers to the e-corder
From eDAQ Wiki
Revision as of 20:43, 5 June 2017 by Oliver (Created page with "* Refer to the manual for the amplifier, which can be downloaded from [https://www.edaq.com/edaq-product-manuals here]. The manuals included detailed instructions on connectin...")
- Refer to the manual for the amplifier, which can be downloaded from here. The manuals included detailed instructions on connecting to the e-corder with diagrams.
- Check the I2C communications cable (the grey cable with 9 pin connectors on both ends) is connected to both the amplifier and the e-corder.
- Verify correct BNC to BNC cable connections between the e-corder input and BNC on back of amplifier; some front ends require more than one BNC to BNC cable.
- If the amplifier has an Instrument Connection Port Adaptor (EC025) at the back of it, make sure adaptor bar is pushed fully into the hardware unit.
- Some amplifiers require external power, such as the ER125 C4D Detector, ER825 C4D multi-channel Detector and EA362 Dual Picostat.